ENRAF-NONIUS ATG Interface. Tank Gauging Protocols Manual. ENRAF INTRODUCTION. The interface between BBI’s Network // Honeywell Enraf ATG Pdf User Manuals. View online or download Honeywell Enraf ATG Installation Manual. Honeywell Enraf series ATG servo gauge is a reliable, versatile and accurate automatic tank level MB; Honeywell Enraf Servo Gauge ATG Manual.
|Published (Last):||21 October 2008|
|PDF File Size:||4.20 Mb|
|ePub File Size:||13.85 Mb|
|Price:||Free* [*Free Regsitration Required]|
Each atv must have a unique address, and hence TA must be programmed differently. Check the correct orientation of the gauge with respect to the tank. The displacer may not enter the nozzle.
EC declaration of conformity Refer to the EC declaration of conformity, shipped with the instrument. The dip mode is cancelled when one of the interface measurements is selected.
Loosen do not remove the Allen key screw and turn the transport bracket the opposite way. Very reliable, versatile and In dip mode the displacer is positioned at distance DH above the product and after a time interval DT, the product is dipped. Item Name Description FR Freeze The displacer remains in its position even when the level reaches the displacer position. Install an explosion-proof mains switch in the mains supply cable to each ATG. Recommended programming sequence The ATG is already pre-programmed at the factory.
If a quad cable is used and all four cores are twisted together, use two opposite cores for Honeywell Enraf field bus lines and the two others for mains. Be aware that measuring wire is unrolled from the measuring drum. The following items contain the error codes of the processor boards: Additional information Please do not hesitate to contact Honeywell Enraf or its representative if you require additional information. Installation guide dedicated O-ring ATG Honeywell Enraf field bus lines may be interchanged.
The displacer is suspended from a strong, flexible measuring wire which is stored on a precisely grooved measuring drum. Warning Make sure that all power to the instrument is switched off before opening the cover of the electronic and terminal compartment.
Type of item Description Commands These will force the gauge to execute a special task or function. Some of the data items are read-only. Refer to figure Note: If initializing of all processors is successful, the display will switch to its default display format, amnual is programmable with item DF.
The warranty terms and conditions applicable in the country of purchase in respect to Enraf B. EX Exit Exit protection level 2.
Honeywell Enraf ATG Manuals
For this purpose the covers of the servo, drum compartment and terminal compartment are fitted with O-rings refer to figure 5.
Additional information Amnual not hesitate to contact Honeywell Enraf or its representative if you require additional information. Raise the displacer to maximum height, at least above the ball valve, when present.
In dip mode, the displacer is raised above the product over the dip height distance DH. Only applicable with XPU Remove the displacer from the measuring wire and attach the smallest test weight of 25 grams.
The possible display formats are: Simple audit trail functionality is implemented by recording several SAT related parameters. AR Accept reference By giving this command, the level value entered in item RL, is accepted as product level and will be shown on the display.
The design of mnual ATG is such that it makes replacement and service simple. For other applications contact Honeywell Enraf. That should be done in this stage.
Enraf servo ATG install guide |
It may damage the electronic circuits. CA Caution If a 45 mm displacer is used, stop immediately the Calibrate command as soon as the displacer is above the ball valve, followed by an FR freeze command.
Open the drum cover. Figure 2 Enra of the gauge on a tank top view On a stilling well or guide pole, the orientation of the gauge may be chosen freely.
Avoid possible generation of static electricity. Do not open any of the instrument covers while power is still connected. On product level the displacer is immersed less. It is recommended to use the service tool to make a log file of the instrument. Check whether the pre-programmed displacer area is in accordance with the used type of displacer. When the XPU detects an error about a certain item, that item follows the error code, separated by a maunal.
Check that non-used cable inlets are sealed with appropriate stopping plugs. Remove the GPS board. AR By giving this command, the level value entered in item RL, is accepted as product level and will be shown on the display.
Honeywell Enraf 854 ATG Manuals
The display will show a BL block or FR freeze in the lower right corner of the display. The transmission ejraf becomes: Be aware that measuring wire is unrolled from the measuring drum. The lock test function is cancelled when the magnet is removed.